Biosensor for determining average iron content of ferritin by measuring its optical dispersion
Gupta, Ruchi; Alamrani, Nasser A.; Greenway, Gillian M.; Pamme, Nicole; Goddard, Nicholas J.
Nasser A. Alamrani
Gillian M. Greenway
Nicholas J. Goddard
© 2020 SPIE. Average iron content of ferritin has a potential to serve as a biomarker for early identification of high-risk trauma patients at point-of-care (PoC). Appropriate therapies can then be administered to reduce morbidity and mortality. Currently, protein and iron levels are measured separately using enzyme-linked immunosorbent assay (ELISA) and UV or atomic absorption spectroscopy (AAS) respectively, but the use of two completely different methods adds to the complexity and analysis time of the combined measurement. As a result, these methods are unsuitable for PoC analysis. To address this gap, we report a biosensor for measuring the average iron content of ferritin in a single step. The biosensor was based on a dye-doped leaky waveguide (LW), which operates in the entire visible wavelength range, and hence allowed the measurement of differences in the optical dispersion of ferritin and apoferritin to determine the average iron content of the protein. The LW biosensor comprised a 1.54 micron thick mesoporous chitosan slab waveguide with immobilized antibodies against ferritin/apoferritin to measure the optical dispersion of 110 nM protein. Based on the baseline noise, the limit of detection for this method is ∼700 pM for ferritin/apoferritin. The biosensor has a significant potential for PoC measurement of the average iron content of serum ferritin and, in future, the total protein cencentration.
Gupta, R., Alamrani, N. A., Greenway, G. M., Pamme, N., & Goddard, N. J. (2020). Biosensor for determining average iron content of ferritin by measuring its optical dispersion. Proceedings of SPIE, 11235, https://doi.org/10.1117/12.2542162
|Journal Article Type||Conference Paper|
|Acceptance Date||Jan 1, 2020|
|Online Publication Date||Feb 21, 2020|
|Publication Date||Feb 21, 2020|
|Deposit Date||Dec 8, 2020|
|Publicly Available Date||Jan 20, 2021|
|Journal||Proceedings of SPIE|
|Publisher||Society of Photo-optical Instrumentation Engineers|
|Peer Reviewed||Peer Reviewed|
|Keywords||Waveguide; Biosensor; Chitosan; Mesoporous; Optical dispersion; Ferritin; Point-of-care; Microfluidic|
© (2020) COPYRIGHT Society of Photo-Optical Instrumentation Engineers (SPIE).
You might also like
Electrochemical immunoassay for the detection of stress biomarkers